Writing Reports is Important for the Floor Installer
Checking out floor problems can be a daunting task. Problems may be very obvious (picture above) but the cause could be hard to determine. The floor may be cupped, may have gaps or other defects, but nobody knows what happened. The floor looked great at the time the installation was finished.
An inspector is called in to find the cause for the floor failure. Accusations fly and the floor installer may be blamed for doing a bad job.
Now is the time for the floor installer report to be presented. The condition of the floor when the installation was finished needs to be evaluated. Not only does the installer report protect the floor installer from unwarranted claims, the installer report allows the inspector to find the cause of the problem by comparing measurements obtained now with measurements obtained at the time of the installation.
Dual-depth meters from Lignomat are ideal to document floor conditions: Readings can be obtained at 1/4″ and at 3/4″ deep. The 1/4″ depth reads the top layer, the 3/4″ depth includes the core. Measurements taken at 1/4″ and 3/4″ deep truly pin-point the moisture condition of the floor.
Thermo-Hygrometer or Datalogger BL2 are ideal to document ambient conditions: Use a Thermo-Hygrometer to measure the relative humidity and temperature at the time the moisture readings were taken. Better yet leave a datalogger BL2 on site to record relative humidity and temperature. The BL2 takes 6 readings per day for 8 years. The BL2 graph actually shows EMC.
A installer report of moisture measurements should include several measurements from random locations and also from moisture sensitive areas such as below big windows, in entrance ways, in kitchens, by the sink and by the refrigerator. Note the following for each measurement:
- Address, name of customer, date
- Type of floor, wood species, manufacturer of floor and supplier
- Name of manufacturer of meter
- Name of moisture meter, measuring mode: pin or scan
- Settings for wood species and temperature corrections (pin mode only)
- Settings for measuring depth (scan mode only)
- Measuring depth for pin, integral pins or external electrode
- Exact locations where the readings were taken (photo can be helpful)
- Moisture value
- Add relative humidity and temperature readings
The photo should show an easy to identify “landmark”: Here a corner with a low window sill and special moldings. A true measurement should show the moisture value and the meter settings if possible. It is also important to show the hand pressing the meter against the floor. (When taking a reading pinless meters should be pressed slightly against the surface.)